measurement of gamma ray absorbed dose level in air in conditions of emergency and post-accident including accidents beyond design basis.
is applied by monitoring service at sites with atomic power plants as well as at other sites related to obtaining, reprocessing and using of radioactive materials.
|Detector type||Ion chamber|
|Measurement range, Gy/h||1,0*10-4 – 1,0*105|
|Energy range, MeV||0,06 – 8,0|
|Working temperature range, oС
for detection unit;
|From 0 to +220
From 0 to +60
|Irradiation stability for detector block and standpipe, Gy/h||106|
|Input current, VA||50|
|Time to failure, h||30000|
Standpipe with access line
Measuring unit BI-43R
|Output type||Encoded in RS-485 interface over two independent channel|